Ambient+Weather+Rain+Gauge+Reads+0.00+after+it+rains

Problem
Ambient Weather Rain Gauge reads 0.00 after rainfall.

Discussion
This is usually a result of a clogged filter or insects nesting in the mechanism, or the thermo-hygrometer-transmitter connector needs cleaning.

Solution

 * 1) Disconnect the phone connector to the thermo-hygrometer-transmitter and clean the connectors with water and a q-tip.
 * 2) Remove the funnel by pressing the two compression tabs on the sides of the rain gauge to expose the tipping mechanism.
 * 3) Remove any bugs or spider webs that may be interfering with the tipping mechanism.
 * 4) Clean the rain gauge filter with a damp cloth and clean the tipping mechanism. Spray with bug spray to prevent future issues.
 * 5) Make sure the connector between the thermo-hygrometer-transmitter and the rain gauge is clean. Clean with cotton swab.
 * 6) If the unit has failed, we can replace the rain gauge under warranty. To get started, please visit: [|www.ambientweather.com/rma.html]
 * 7) If out of warranty, you can order a replacement here: []