Rain+Rate+reading+incorrectly

Problem
My rain rate is reading incorrectly.

Discussion
Virtual Weather Station either obtains the rainfall rate from the console or calculates the rainfall rate. To calculate rain rate, Virtual Weather Station determines the current total rainfall as reported by the console, and then determines the total rainfall one hour ago from the database. The database values can be viewed by selecting Settings | Database Settings. Recall the last record to review the last record written to the database.

Solution
1. If you recently changed your total rain, the rain rate will be affected. 2. If your weather station clock is out of sync with your computer clock, and you own a logging station (such as a Davis Instruments unit), it will not properly calculate the rainfall one hour ago. Make sure your computer and weather station clocks are synchronized. 3. If there is a bad date record written your database, the rain rate will always read a non-zero value. To review this, select Settings | Database Settings and review the last record on the database (select Recall Record No:). If the date is some time in the future, delete this record. Keep deleting all records until the date and time is less then the current date and time. If this date is in the distant past, correct the time on your weather station clock.