WeatherLink+USB+Communication+Issues+and+Virtual+Weather+Station

Problem
WeatherLink software communicates fine but Virtual Weather Station does not.

Solution
> The problem is related to the DLL that Ambient Weather utlizes and is written by Davis Instruments. This DLL does not allow comport assignments greater then Port 9. The WeatherLink software does not have the same problem because Davis does not use the DLL, they use direct communication protocols. The reason we use the DLL is it is much more reliable then their direct communication protocal. The direct communication protocal has a number of band-aid fixes that only Davis is aware of, and makes writing softwar cumbersome. > If you unistall and reinstall the USB driver software, in general, this will reassign the comport to 1-9 and allow the DLL to work.
 * 1) Both Virtual Weather Station and WeatherLink can not communicate to the same communication port at the same time. Make sure only one program is running at the same time.
 * 2) Make certain the comport and baud rate settings are identical for both programs.
 * 3) If you are using the WeatherLink version of the data logger, USB driver software was included. If the driver software assigns a communication port greater then 9, you will have to reinstall the drivers.