USB+Device+Stopped+Communicating,+WMR-100

Problem
My WMR100, WMR200, WMR88 USB (or USB converter) interface stopped communicating to Virtual Weather Station.

Solution
> http://www.ambientweather.com/do30tr.html
 * 1) **Power down and power up the display console** by disconnecting from the AC adaptor and removing the batteries. Wait 10 seconds and power the unit back on.
 * 2) Make certain you are running Version 14.00p54 or greather. There were changes to the comport handling to prevent drop outs. To determine what version of software you are running, select Help | About Virtual Weather Station. To download the latest version, please visit the following link:
 * 1) Make sure your date and time is set properly on the console. Failure to set a correct date and time will result in values being ignored. It must be within one year of your computer date and time. If you have modified the date and time on your computer, it is recommended you correct this.
 * 2) If you are connecting through a USB hub as opposed to direct USB connection, connect directly to eliminate issues with the hub.
 * 3) Turn off all screen saver applications, which can interrupt USB data.
 * 4) Turn off all power save options. To view your options, select the **Windows Control Panel**, **Power Options** and select **Never** for all of the power schemes. Select the Hybernate Tab and make sure the hybernate checkbox is unchecked.
 * 5) Turn off all USB power save features. Select the **Windows Control Panel**, **System**, **Hardware**, **Device Manager**, **Universal Serial Bus controlllers**, and select the plus (**+**) sign to view all of the USB controllers. You should see several **USB Root Hub** entries. **Right click** on each entry (unless you know which one is correct), select **Properties**, and under the **Power Management tab**, **uncheck** the selection to **Allow the computer to turn off this device to save power**.
 * 6) Select **Communication** | **WMR100 Sensor Communication** from the Virtual Weather Station menubar. If any of the LEDs are not green but checked, uncheck the inactive sensors, they are currently not reporting to your weather station. If specific sensors are not reporting to the console, please contact Oregon Scientific Customer Service for additional support at 800-853-8883.
 * 7) If the problems persists, try another computer.
 * 8) If the problem continues, contact Oregon Scientific Customer Service for additional support at 800-853-8883.